TITLE

# Measurements of total peroxy and alkyl nitrate abundances in laboratory-generated gas samples by thermal dissociation cavity ring-down spectroscopy

AUTHOR(S)
Paul, Dipayan; Furgeson, Amanda; Osthoff, Hans D.
PUB. DATE
November 2009
SOURCE
Review of Scientific Instruments;Nov2009, Vol. 80 Issue 11, p114101
SOURCE TYPE
DOC. TYPE
Article
ABSTRACT
A novel measurement technique, thermal dissociation cavity ring-down spectroscopy (TD-CRDS), for rapid (1 s time resolution) and sensitive (precision âˆ¼100 parts per trillion by volume (10-12; pptv)) quantification of total peroxy nitrate (Î£PN) and total alkyl nitrate (Î£AN) abundances in laboratory-generated gas mixtures is described. The organic nitrates are dissociated in a heated inlet to produce NO2, whose concentration is monitored by pulsed-laser CRDS at 532 nm. Mixing ratios are determined by difference relative to a cold inlet reference channel. Conversion of laboratory-generated mixtures of AN in zero air (at an inlet temperature of 450 Â°C) is quantitative over a wide range of mixing ratios (0â€“100 parts per billion by volume (10-9, ppbv)), as judged from simultaneous measurements of NOy using a commercial NOâ€“O3 chemiluminescence monitor. Conversion of PN is quantitative up to about 4 ppbv (at an inlet temperature of 250 Â°C); at higher concentrations, the measurements are affected by recombination reactions of the dissociation products. The results imply that TD-CRDS can be used as a generic detector of dilute mixtures of organic nitrates in air at near-ambient concentration levels in laboratory experiments. Potential applications of the TD-CRDS technique in the laboratory are discussed.
ACCESSION #
45515524

## Related Articles

• On the Application of Cavity Ringdown Spectroscopy to Measurements of Line Shapes and Continuum Absorption. Cormier, John G.; Hodges, Joseph T.; Drummond, James R. // AIP Conference Proceedings;2002, Vol. 645 Issue 1, p401

Cavity ringdown spectroscopy (CRDS) is a highly sensitive spectroscopic technique that has been successfully applied to problems such as trace gas detection and the observation of weak spectra. Despite possessing several intrinsic advantages over other techniques, CRDS has not yet been widely...

• Low temperature cell for cavity ring down absorption studies. Lewis, Ernest K.; Moehnke, Craig J.; Navea, Juan G.; Manzanares, Carlos E. // Review of Scientific Instruments;Jul2006, Vol. 77 Issue 7, p073107

Phase shift cavity ring down is a technique that due to its long optical path length is an ideal method to detect weak absorptions. Coupling the method to a custom fitted cryostat allows gas phase molecules to be studied at cryogenic temperatures in a thermally isolated vacuum chamber. A novel...

• Configuration of ring-down spectrometers for maximum sensitivity. Jakubinek, Michael; Tong, Zhaoguo; Manzhos, Sergei; Loock, Hans-Peter // Canadian Journal of Chemistry;Jun2004, Vol. 82 Issue 6, p873

Cavity ring-down (CRD) spectrometers used for analytical applications frequently have design requirements different from spectrometers used for gas-phase spectroscopic applications. A formalism that allows for maximization of the relative sensitivity by adapting the cavity length and absorption...

• Theoretical analysis on the dynamic absorption saturation in pulsed cavity ringdown spectroscopy. Lee, J. Y.; Hahn, J. W. // Applied Physics B: Lasers & Optics;Sep2004, Vol. 79 Issue 5, p653

We perform a theoretical analysis on the transient dynamics of absorption saturation in pulsed cavity ringdown spectroscopy. Based on a coupled rate equation approach, modelling of the dynamic saturation is carried out to account for time-evolving intracavity photon density and absorbing...

• Absorption spectra of lithium dimer in the region of X[sup 1]Î£[sub g][sup +]-B[sup 1]Î [sub u] band by cavity ringdown spectroscopy. Labazan, Irena; Milosˇevic´, Slobodan // AIP Conference Proceedings;2001, Vol. 559 Issue 1, p307

We present cavity ringdown spectra of lithium vapor generated in the heat-pipe oven. By closer inspection of absorption in the wavelength range corresponding to two-photon atomic Li(2s-3d), Li(2s-5d), Li(2s-6d) and quasiresonant Li(2p-4s) transitions we examine which of them could be candidates...

• Measurement of multi-exponential optical decay processes by phase-shift cavity ring-down. Bescherer, K.; Barnes, J. A.; Dias, S.; Gagliardi, G.; Loock, H.-P.; Trefiak, N. R.; Waechter, H.; Yam, S. // Applied Physics B: Lasers & Optics;Jul2009, Vol. 96 Issue 1, p193

Multi-exponential decay waveforms are common occurrences in cavity ring-down spectroscopy and the respective ring-down times are typically obtained by fitting the ring-down waveform to the sum of exponential decay functions. In phase-shift cavity ring-down (CRD) spectroscopy the measurement of a...

• Infrared spectra of the CF3I dimer: A concurrent application of matrix-isolation spectroscopy and cavity ring-down spectroscopy. Ito, Fumiyuki; Hirabayashi, Shinichi // Journal of Chemical Physics;6/21/2006, Vol. 124 Issue 23, p234509

We have observed infrared spectra of the CF3I dimer produced in a supersonic jet by matrix-isolation Fourier transform infrared spectroscopy and infrared cavity ring-down (IR-CRD) spectroscopy. In the matrix-isolation experiments, the dimer was isolated in an Ar matrix by the pulse-deposition...

• Single-cell detection by cavity ring-down spectroscopy. Tarsa, Peter B.; Wist, Aislyn D.; Rabinowitz, Paul; Lehmann, Kevin K. // Applied Physics Letters;11/8/2004, Vol. 85 Issue 19, p4523

The implementation of cavity ring-down spectroscopy in an optical fiber resonator extends the viability of this highly sensitive technique for label-free detection of biological species. By chemically treating the surface of discrete tapered sensing regions along the length of a physically...

• High-resolution cavity ring-down spectroscopy measurements of blended H2O transitions. Lisak, D.; Hodges, J.T. // Applied Physics B: Lasers & Optics;Jul2007, Vol. 88 Issue 2, p317

We probed four closely spaced rovibrational water vapor absorption transitions near $\tilde{\nu}$ =7100 cm-1 using frequency-stabilized cavity ring-down spectroscopy. Room-temperature spectra were acquired for pure water vapor in the Doppler limit and for mixtures containing â‰ˆ6.6 Î¼mol...

Share